|Date Added:||25 March 2007|
|File Size:||30.64 Mb|
|Operating Systems:||Windows NT/2000/XP/2003/2003/7/8/10 MacOS 10/X|
|Price:||Free* [*Free Regsitration Required]|
Most Agilenh products have PDF manuals that can be found on product pages. Please consult your instrument’s Service Guide or contact your local Keysight service center.
DSOA Oscilloscope, MHz [Obsolete] | Keysight (formerly Agilent’s Electronic Measurement)
When oscilloscope manufacturers specify vertical accuracy, they typically only mention bandwidth and DC accuracy. All current Keysight scopes with TV triggering are capable of exclusively triggering on odd or even fields.
Contact an Expert Chat Live. Most Keysight oscilloscopes give 8-bit vertical resolution at normal acquisition mode.
DSO3202A Oscilloscope, 200 MHz
Obsolete View Service Options.
The bandwidth specification addresses the AC accuracy. How can I use a Keysight oscilloscope as a basic I-V curve tracer? When a probe is connected, the scope changes its attenuation setting so that it correctly displays the voltage coming from the probe.
The DSO Series oscilloscope hangs at the first screen after power up and does not boot to the oscilloscope default page. How do I update the Scope Connect software and Series oscilloscope firmware?
What is agielnt amplitude accuracy of my oscilloscope? What options do the DSOseries oscilloscopes offer for saving data? Tighter AC accuracy specifications are very uncommon.
Agilent DSO3202A 200 MHz 2 Channel Digital Oscilloscope
What can I do? No, you must be a “Standard user” or an “Administrator” Where can I find manuals for discontinued and obsolete Hewlett-Packard, Agilent and Keysight test equipment? Can I program a waveform captured by my oscilloscope into a Keysight arbitrary waveform generator? Other vendors also supply old manuals.
When using TV trigger mode with a Keysight oscilloscope, can I trigger only on even or odd fields? You must remove the device driver from Device Manager and cycle power on the oscilloscope so that the proper USB driver is installed.
How often should I send in my oscilloscope for calibration? Why do I see more noise on my oscilloscope when a probe is connected? You can save data ahilent internal oscilloscope memory or you can use the NA Scope Connect software to save data to your PC.
Please disconnect and re-connect the NA communication module